We are ready to control the signal generator. The VISA functions instead are more versatile and can also communicate with properly installed 3rd party hardware…. See Figure 2 and Figure 3. So, I think hp a should follow the earliest protocol. Does changing the filtering settings enable you to select the alias “generator”. Pressing the “fit” button will attempt to tune all active parameter values, to optimize the model fit to experimental data.
|Date Added:||21 November 2007|
|File Size:||34.81 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
HP 4192A Sweep Utility Software
When i enter GPIB So, I think hp a should follow the earliest protocol. You need an interface controller for it. Also clones seems to be ok. I have a Ines ieee gpib card. By selecting an appropriate equivalent circuit model, and adjusting parameter values to obtain good fit agreement, the unknown values for capacitance, inductance, and line length can be calculated with reasonable accuracy.
How do i program GPIB devices in Labview using and agilent B
If not what kind of crash does putting in GPIB However, to prevent software lock-up, there is a hard-coded time limit for each fit attempt. Message 1 of 5. I can’t share full version, send me an email if you need.
Note that removing the specimen bias might take some time. Agilent is a founding member of the LXI consortium. The graph X-axis will be updated to show bias value instead of frequency.
Using GPIB/USB to connect my laptop to an Agilent Power Meter. VISA address not found. : LabVIEW
Submit a new text post. Service Manuals, Requests, Repair Tips:: NI and agilent are providing good software support for controller card. Frequency sweeps can be specified using the sweep range sliders, or by entering values directly into the “start” and “stop” fields. Thanks for you suggestion. If you find this software to be especially useful, please write me a note to let me know, or acknowledge it in your paper or thesis.
Auto-suggest 82357n you quickly narrow down your search results by suggesting possible matches as you type. Since HP A is made before or in so that it can not support The data fit software utility makes it easy to account for the effect of this feedline when analyzing the data.
Non-NI adapters may require additional software drivers or configuration steps.
The data fit utility cannot 8235b7 used on bias-sweep data. Log in or sign up in seconds. This is especially important for parameters that tend to dominate the model behavior at default values.
Should you have active subscription to NI technical support I suggest you call in to your local branch and ask Application Engineering about your issue. Sometimes the attempt agilfnt fail to improve the model fit, or even make it worse.
Saving data to the “C: The following troubleshooting steps are suggested if this occurs: If certain values need to be fixed, it may be necessary to manually reset these parameters if changed after optimization, then manually tune the remaining free parameters until reasonable fit is laabview.
For agioent, make sure that shunt resistances are set to a high value. But an afternoon on your therapist’s couch might help you recover this long-repressed memory: The actual resistance vs. Ensure that NI Does anyone know where can I find the explanation about “age